In thermocouple could we use the lug for termination or
not,Thermocouple is any time,I am currently concerned with
K type.please let me know if anybody know
Answers were Sorted based on User's Feedback
Answer / jigaranadkat
LAW of intermediate metal comes into picture, according to
which resulting temperature will be the algebric sum of all
EMF produced inbetween.
But with LUGs perticularly, the temperature difference will
be very less between so less almost negligible amount of
EMF will be produced. Hence NO HARm.
In order to achieve high accuracy it has been observed that
perticularly in ares where temperatures are TOO HIGH,
LUGing is to be avoided.
|Is This Answer Correct ?||11 Yes||0 No|
Answer / george mathews
The basic principle behind any Thermocouple is that there
are two disimilar material connected at one end that is
called the Hot Junction , and we take our out put from the
other open junction that is the cold junction. But when we
connect a lug there is anather metal thats mean 3 metal and
we ll definitely get a wrong out put and in a high
temperature area or zone this may make a big difference.
|Is This Answer Correct ?||14 Yes||5 No|
Answer / bm
Dont see any harm in connecting lug for termination
provided the continuity remains the same i.e there will be
very very less resistance drop if proper connection is not
there with thermocouple wire, lug and the terminals.
|Is This Answer Correct ?||6 Yes||5 No|
Answer / rathina
Normal practice, Leg connection not recommended
|Is This Answer Correct ?||1 Yes||2 No|
While giving Power Supply of 24 VDC to a Transmitter,why Shield Terminal is connected either at Transmitter or at ISB.Why not Shield Terminal connected at both ends.If the Reason is to avoid Noise interference,which type of noice that is interfering.Is the Noice from the 24 VDC supply terminals or from any external source.
can anybody tel me how a vfd or any field instrument communicates/interface with plc/scada/cs.
Functions of electropneumatic valve positioner:
what is difference between RTD and Thermocouple?
Why any transmitter signal is 4-20 mA only Why is started from lower range 4 mA and upper range is 20mA.
pls tell me whethere flame proof HMI's are available or every HMI is manufactured by considering flame proof standard.
what is the deference bitween DCS and PLC SCADA HMI System for Power Process Industry.
what is modbus,profibus,fieldbus
how to calculate MMWC values in to Tonns(MMWC values showing in DP transmitter and Tonns value showing in Totalizer). please tell me details?
HOW FOREIGN CONTROL SYSTEM CAN BE CONNECTED WITH DCS?
what are turbo generators?
What is transducer