What is the working principle of oxygen analyzers(zirconium oxide type) used in CFBC boilers?
Answers were Sorted based on User's Feedback
Answer / suresh xavier
there are two parts one is the sensing part and other is
converter. sensing part contains zirconium and heating
mechanism. the main caractristic of Zr. is at higher temp.
(= 750 deg.) the proprty of Zr., it allows the O2 ions to
conduct. to maintain the Zr. at 750 deg. normally we use
platinum electrode on the both side to create a heating
mechanism. when the probe is placed in the flue gas path one
side of the probe will maintaine the span gas (instrument
air contains 21% O2) and the other end is in the flue gas
path contains unknown % of O2. thus it will create a diff.
partial O2 on the each side of the cell. electrode in
contact with higher O2 pressure act as -ve electrode and it
will create an emf. the coverter pck that emf and using the
calibration produce the % O2.
for zero calibration use zero gas (0.98% O2 ~~ nitrogen gas)
for span we use 21% O2 (instrument air)
|Is This Answer Correct ?||32 Yes||2 No|
Answer / arsi vadher
what is out in MV and % of oxyzen when on Both side pressure is Atmospheric?
|Is This Answer Correct ?||5 Yes||0 No|
Answer / navneet jain
900 mv and oxygen 0 %
|Is This Answer Correct ?||0 Yes||0 No|
principle of automatic votage regulator
hi am durga, i just now completed my eie  , can anyone please guide me what r courses i need to join to enter into core company
In instrumentation standard we use 4 to 20ma for transmission , why cant we use 3 to 19ma ?
List three types Level Transmitters with their working principles
Explain plc level and its type?
components of control circuit belongs to control valve
what is plc?
what is the difference b/w magnetic flow meter&mass flow meter? which one is best?
What is diffrence between Transducer and Transmitter?
wht is Zero Elevation and Zero Suppression?
what is Least Value of MICROMETER?
What is cable size for intrenal wiring in DCS or PLC ,from any card to Terminal amd Why