Follow Our FB Page << CircleMedia.in >> for Daily Laughter. We Post Funny, Viral, Comedy Videos, Memes, Vines...

Define magnetic flow meter and mass flow meter.

Define magnetic flow meter and mass flow meter...

Mass Flow Rate represents the amount of mass that passes a
specific point over a period of time.

Mass flow rates are used to measure the weight or
mass of a substance flowing through a process
operation. If the volumetric flow rate and density
are known, the calculation is based on the formula

W = Q x r
where
W = mass flow rate
Q = volumetric flow rate
r = density (r = density “rho” )
Megnetic flow meter:
What is a Magnetic Flow Meter?

A magnetic flow meter (mag flowmeter) is a volumetric flow
meter which does not have any moving parts and is ideal for
wastewater applications or any dirty liquid which is
conductive or water based. Magnetic flowmeters will
generally not work with hydrocarbons, distilled water and
many non-aqueous solutions). Magnetic flowmeters are also
ideal for applications where low pressure drop and low
maintenance are required.

Principle of Operation
The operation of a magnetic flowmeter or mag meter is based
upon Faraday's Law, which states that the voltage induced
across any conductor as it moves at right angles through a
magnetic field is proportional to the velocity of that
conductor.

E is proportional to V x B x D where:

E = The voltage generated in a conductor
V = The velocity of the conductor
B = The magnetic field strength
D = The length of the conduc

 Is This Answer Correct ? 27 Yes 1 No

More Instrumentation Interview Questions

we have LPG spherical Tank and we want to install DP Xmtr for level measurement.below are the physical measurements Total height of tank is 1850 mtr but 0% level is 1000mm above bottom and 100% level is 450 below upper side.What is the LRV and URV and please mention the formula .The SG of LPG is 0.53 and Wet Leg method will be used.

How to calculate 47Liter H2 Gas cyilnder consumtion? Cylinder inlet Pressure is 200bar and Discharge pressure is 6.0bar. Flow is 100ml/min.For the above condition how many days one cylinder can be used?

why we set -10 volt in vibration sensor output,why not +10 volt

sir, i preparing hpcl,ongc,barc,hal exams(instrumentation engg)pls send a placement papers

To find closed tank level by DP transmitter , how to findout its range.and insomeplace they using range like - 1500 to +1500 mmwc why not 0 to 3000 mmwc. how they spilit they range for minus and plus.and why?

In bimetallic temp. Gauges why does metal strip bend? Why do we use low expansion factor metal and high expansion factor metal in bimetaliv strip?

How to calculate P,I&D value in PID controller ?

the basic of a DP flow transmitter are the following

Dear Sir/Madam,im going to attend Hpcl written exams on feb 17th in instrumentation stream..can anyone send me the HPCL model papers and pattern of HPCL ....i ve not yet attended anything in HPCL...so i dont know how it comes...so just i need some model test papers and pattern ..pls help me friends...

How to calculate the flow rate in Vortex flow meter?What is K factor?Can you give me an example calculation to understand the same as better?

What is the syllabus for Instrumentation Engineers, who wish to join the Public Sector Units??

Categories
• Civil Engineering (5076)
• Mechanical Engineering (4445)
• Electrical Engineering (16607)
• Electronics Communications (3915)
• Chemical Engineering (1092)
• Aeronautical Engineering (214)
• Bio Engineering (96)
• Metallurgy (361)
• Industrial Engineering (258)
• Instrumentation (2992)
• Automobile Engineering (332)
• Mechatronics Engineering (97)
• Marine Engineering (123)
• Power Plant Engineering (172)
• Textile Engineering (575)
• Production Engineering (0)
• Satellite Systems Engineering (106)
• Engineering AllOther (1378)