How can we calibrate pressure and flow transmitter in the
Answers were Sorted based on User's Feedback
This procedure can be used to calibrate any type of
instrument in the field on a live plant regardless of it's
type or application. The only variation in the procedure is
what test equipment you will use to calibrate the instrument
with. For instance calibrating a temperature transmitter you
want to take a HART and decade box with you. Calibrating a
pressure , flow or DP transmitter, you want to take a HART
and pressure pump with you like a Druck or hand pump with a
precision gauge on.
Since the transmitter is on line and on a running plant, you
need to determine a couple of things first before you start
working on it. First and most important of all, what does it
do? Is it a control transmitter, a trip transmitter or just
Control transmitter means it's output will go to the DCS and
is linked to a PID controller which will give a continues
output, based on this input, to a control element in the
plant plant like a control valve or a heater element. These
are all critical control components that will trip the plant
if you upset their stability.
Trip transmitter means that it is linked to the ESD system
and will start a plant shut down if it is seeing a to high
or low pressure, flow, DP or level depending on the
application you are working on. Just indication means that
it is completely safe to work on and you cannot trip anything.
To make the control transmitter safe to work on you need to
put the DCS PID controller in manual. This will keep the
output of the controller at the current output position and
bypass the output function of the controller. In a fast
moving application the control room operator will send a
field operator with you to control the control valve
manually with it's hand wheel while you work on the control
To make the trip transmitter safe to work on you need to put
a inhibit in the ESD system output trip function of the
transmitter. This is also sometimes refer to as a MOS
(maintenance override switch).
In both cases the control room operators will do this for
you providing that you have all the permit to work and
inhibit permission paperwork in place.
|Is This Answer Correct ?||34 Yes||10 No|
Answer / pangavenkatesh
How to calibreate dp transmitter
|Is This Answer Correct ?||8 Yes||0 No|
Answer / anjal
first close the isolation and open the equaliser if its
show zero ok other wise calibrate the transmitter
|Is This Answer Correct ?||18 Yes||12 No|
What should be hight of CONTROL panel for access & OPERATION What is BIS & INDIAN STD for SAME?
difference between dcs & pld
how can we calibate ph meter. how can we calibrat conductivimeter how can we calibrat massflowmeter & it s working.
why we use only 4-20 ma signal as AI and not 0-20 ma?
How does a redundant controller comes in line when the main controller fails??
what is the equation for gas flowmetering? whic includes gas flow, pressure, and temperature.
T.x range -0.1 to 0.2 milibar,so how much mili ampear at 0.1 milibar?
Regarding MOS (maintenance override switch), I want to know its function and its state ( 1 or 0 when not activated ). An example will be very helpful..thank you.
wich is coming the first HAZOP OR SIL
Sir i am studying M.sc,(Tech.)Electronics and Instrumentation in Andhra university.What is the offers for me?
Convert a pressure, level, flow, or temperature process variable value into the appropriate pneumatic or electrical value given different types of instruments.
How does a single acting, spring return ball valve actually work? Does it require continuous air supply?