Follow Our FB Page << CircleMedia.in >> for Daily Laughter. We Post Funny, Viral, Comedy Videos, Memes, Vines...

How to Calibrate variable Head flowmeter (Orifice), here we
dont know High pressure and low pressure then how to apply
pressure on HP and LP side ?

How to Calibrate variable Head flowmeter (Orifice), here we dont know High pressure and low pressu..

The situation must be an installed, running system, with a receiver that displays a flow rate from the DP pressure signal, because no one just has a DP transmitter across an orifice plate for fun.

Every differential pressure primary flow element is assumed to be zero flow at zero DP. So the DP low side pressure 'zero' will always be zero DP, or when the 3 valve manifold has its process connection valves closed and the equalizing valve open.

The measured DP represents some flow rate on the square rooted curve of the normalized percentage of span.

In order to determine the DP span value, you measure the actual DP and note what the flow rate reads on the receiver (PLC/DCS/recorder/indicator/controller) at that DP.
Then you can calculate what the DP span value is for the max flow rate that receiver is configured for.

The receiver has some configured flow range: 0-500gpm? 0-1,000 bpd? 0-650pm? Find out what it is.

For example, say you measure a DP value of 700mm wc and that 700 mm wc shows a flow rate reading of 264.5 lpm on the receiver.

You have discovered that the receiver is scaled 0-500 lpm for that DP transmitter and the receiver does the square root extraction.

What is the flow rate as percentage of flow span?
264.5 lpm/500 lpm = 52.9% of flow rate span

52.9% normalized is 0.529

Since the flow rate is the square root of the normalized DP span, you need to square the normalized percentage of flow rate to find the equivalent percentage of DP span.

0.529^2 = 0.2798, or 27.98% (~ 28%) of the DP span.
If 700mm is 28% of span, 100% of span is 2500mm wc.

So the DP span for the transmitter is 0 (no flow) to 2500mm wc (max flow).

Carl Ellis
Measure First

 Is This Answer Correct ? 0 Yes 1 No

More Instrumentation Interview Questions

HI friends.I am Surya a BTech in Electronics and Instrumentation Engineering.I am going to appear Relince Industries Ltd all India writen test very soon. so plz help me by sendind the probable questions about my subject.My mail ID is surya.barik@gmail.com.

If you SOURCE your outputs from a PLC, would you have to make the output HIGH, or LOW, in order to switch an output ON?

Why mmwc unit is used?

whats the diff b/w zero pressure and obsolute pressure

Boolean algebra

Why thermo wells are used? What materials are used in thermo wells?

Ega interview going on for power plant. Which type of question will asked. For questions

what is the difference b/w magnetic flow meter&mass flow meter? which one is best?

Conversions

why the SCAN time of DCS is more & that of PLC is less? if the SCAN time of DCS then why we use DCS.

Where is wet leg is used?

WHAT IS A SYMPTOM OF FAULT WITH THE SPAN OF AN INSTRUMENT? 1)ZERO-POINT AND 100% ARE CORRECT BUT NOT A MID-RANGE. 2)ZERO-POINT IS CORRECT ,BUT 1005 INPUT IS NOT GIVING 100% OUTPUT. 3)LINEARITY PRBS. 4)1005 O/p IS CORRECT BUT ZEO POINT IS WRONG.

Categories
• Civil Engineering (5074)
• Mechanical Engineering (4444)
• Electrical Engineering (16594)
• Electronics Communications (3915)
• Chemical Engineering (1092)
• Aeronautical Engineering (214)
• Bio Engineering (96)
• Metallurgy (361)
• Industrial Engineering (258)
• Instrumentation (2986)
• Automobile Engineering (332)
• Mechatronics Engineering (97)
• Marine Engineering (123)
• Power Plant Engineering (170)
• Textile Engineering (575)
• Production Engineering (0)
• Satellite Systems Engineering (106)
• Engineering AllOther (1377)