In Assy by potentio metry titration method how select MET U ,DET U, SET U,
No Answer is Posted For this Question
Be the First to Post Answer
Explain in detail what are the procedure and calculations involved for accuracy (%Recovery)in method validation?
What is the calibration of uv process and preparation
How to select Sample concentration in RS analysis?example.
why should we calibrate pH meter with 4,9.2 and 7.4 buffers particularly?
full name of OOS & OOT ? Dercribe it with details ? why this is needed in pharma industry ?
give clarity of linearity and range in method validation
plz tall me that any pharmacopeia and guideline is par meted to use 1.68,4.01,7.00,9.00,12.45 buffer for ph meter calibration
why results are variation in same samples analysis in gc and hplc ?
Why we more noise in PDA detector compared to UV detector?
formula for %RSD.
What is the principle of hplc detectors like.. UV detector,flurosence detector, RID?
why ph is between 0 to 14