we are used SAMSON 3730 Smart positioner.please tell me how to calibration this.
Answer / chanchal
at first check the air supply.if it is ok then
feed the 4mA by using calibrator in i/p converter of positioner. if valve is zero position(fully closed) then its ok.if valve is not in zero position(fully closed) then set the zero position in positioner by using zero point screw.
next feed the 20mA by using calibrator in i/p converter of positioner.if valve is 100% open then its ok.if valve is not 100% open. then adjust the span setting by using span screwin positioner.
then check 25%,50%,and 75% open position in valve by feeding 8mA,12mA,and 16mA by using calibrator in same way.
|Is This Answer Correct ?||4 Yes||5 No|
Draw a complete P&ID of a flow suction tank indicating all control & protection equipment. Indicate all its inlets and outlets. Explain the controls for this system
If mVolts can drop by resistance of wire as compare to mA, then how thermocouple mV reaches in CCR as accurate value?
What information needs for Control Valve "sizing"
What is DCS?
deference between DCS and PLC including date processing and architacture.
what is the Analog signal and what is the Digital signal
what is maxdna ?. explain about the software details.
what is diffrence between cascade & three element system.
what is effect of specific gravity in differential level transmitter , explain formula related with this .
what is the meaing of calibration curve?
Can Any body tell me how to check Modbus Settings in ER300N L&T trivector meter?
why we use 24 vdc in instruments mostly?