Answer / kishor solanke (alkem)
To check the area(response) of analylite in the sample is
linear with respect to it's concentration OR injection
volume.This parameter done by two method:
1) Detector linearity:
In this linearity test the solution of different
concentration (linear conc.)are injected in triplicate of
each solution & linearity of each concentration calculated
as co-relation coefficient by graphical pattern in excle
against mean srea obtained.
In this test the soluiton of same concentration injected in
triplicate with different injection (linear)volumes &
linearity calculated as above.
|Is This Answer Correct ?||20 Yes||0 No|
What's the importance of these procedures?
reasons for negative peaks in chromatography
What is split
NMR is both qualitative and quantitative?EXPLAIN
Which products will produce when acetic anhydride reacts with Ter-butanol, Sec-butanol, Iso butanol, Dimethylamine etc.
how to calculate corelation coefficient in G.C Calibration? what is the calculation part?
How NMR spectra can be calibrated in presence or absence of TMS?
Is it necessary to validate both assay and related substances for usp official methods for products.
What is the difference between chromatographic purity and related substances?
What is the definition of RRF
how much mass should be there in volumetric flask while in preparation of sample for assay?
What is the difference between drug product and drug substance?