Why we more noise in PDA detector compared to UV detector?
Answer / anil
There is no Integration function in the conventional UV-Vis spectrophotometer which
accumulates the signal. For example, the total required time will be 1000 sec. for 1000 data
points and it takes 1 sec. to measure one datum. In this case, all 1000 data have the same signal
to noise ratio (S/N).
But in a PDA instrument which has a 1000 photodiode array, 1000 data points can be
measured in 1 sec. and it would take 1/1000 sec. to achieve the same result obtainable in 1 sec.
in a conventional instrument. Therefore, when the same sample is measured for 1000 sec in a
PDA instrument, the signal is accumulated and is 1000 times greater than when measuring for 1
sec. The noise will be 10001/2
. This means that the S/N ratio is improved by 10001/2
This resulting benefit of fast data acquisition is termed Felgett’s S/N Advantage or
|Is This Answer Correct ?||10 Yes||1 No|
In case of HPLC coloums what is the difference Between C8 & C18 coloums & how can we select difference type of coloums like Inertial ODS , Hypersil BDS , ODS etc.?
Which type of material analysis in GC using Theramal conditivity detector ?
Why triethylamine and o-phosphoric acid used while setting pH of mobile phase during HPLC Analysis?
Why do we use 0.005M h2so4 solution for preparation of k2cr2o7 solution for UV calibration?
wat is reverse phase in hplc
why salicylic acid not use for dissolution.
What are the differences between GC and HPLC?
what is LC - MS ? What its wide use in cro/pharmaceutical Industry
is it necessary to do HPLC calibration at wavelength 315nm if we are doing analysis at this wavelangth
What is void volume & dead volueme
which procedure we follow if 6th month of accelerated sample fail?
can i use hplc detector to uplc and why?