Why we more noise in PDA detector compared to UV detector?
Answer / anil
There is no Integration function in the conventional UV-Vis spectrophotometer which
accumulates the signal. For example, the total required time will be 1000 sec. for 1000 data
points and it takes 1 sec. to measure one datum. In this case, all 1000 data have the same signal
to noise ratio (S/N).
But in a PDA instrument which has a 1000 photodiode array, 1000 data points can be
measured in 1 sec. and it would take 1/1000 sec. to achieve the same result obtainable in 1 sec.
in a conventional instrument. Therefore, when the same sample is measured for 1000 sec in a
PDA instrument, the signal is accumulated and is 1000 times greater than when measuring for 1
sec. The noise will be 10001/2
. This means that the S/N ratio is improved by 10001/2
This resulting benefit of fast data acquisition is termed Felgett’s S/N Advantage or
|Is This Answer Correct ?||10 Yes||1 No|
Basic principle of ESI?
what is difference between assayand potency
when we use capillary and packed columns in gc
In the isomers,enantiomers give the only physical properties present such that no chemical properties present.
How you will select the runtime in HPLC for a drug in method development.
What is linearity in HPLC and how will you test?
Why dst is used in kf for factor determination less than 1
Why we have to take 42gm of sodium hydroxide instead of 40gm for preparation of 1M Sodium hydroxide as per Pharmacopia? Whether it means the formula for calculating molarity was wrong?
In GC calibration,what is the procedure to calibrate the TCD (thermal counductivity detector)suggest ?
Defination of ZEEMAN EFFECT in NMR ?
Qc interview questions
principle of FTIR