Why do we use KCl ,K2Cr207,Toulene & Halonium oxide for the caliberation of UV
Answers were Sorted based on User's Feedback
Answer / kumar
kcl will be used in Limit of stray light, toluene and hexane will be used in limit of resolution,K2Cr2o7 will be used in control of Absorbance and Holmium oxide will be used in control of wavelength
|Is This Answer Correct ?||20 Yes||5 No|
Answer / dipti shah
to check accuracy of wavelength scan
|Is This Answer Correct ?||2 Yes||13 No|
what is the use of performing tapped density in pharmaceutical industry
What is NPOC method in TOC?
we are performed SOR for a particular product the limit is -6 to -10.We face particular bathes it is not meeting the specification.we performed diffrent instrument diffrent analyst we are getting diffrent diffrent results.solvent is dimethyl sulfoxide what could be the reasion
why should we perform dissolution PVT calibration only by UV spectrometer, not by HPLC ?
Why irmoisture balance is used for corrugated box moisture testing
why chiral separation?
why is salt andv ice used in liquifyiny SO2
how to calculate the Step and ripple accuracy for pump gradient test for UPLC?
Can we do the MC analysis of a sample which is not completely soluble in the medium choosed.
How to Perform a Recrystallization ?
what is the effect of injection volume calculate flow rate in perkin elmer headspace? (ccm parameter in HS)
Why Deuterium lamp is used as UV light Source for UV.