is here anybody who knows how to calibrate the pressure
Answers were Sorted based on User's Feedback
This procedure can be used to calibrate any type of
instrument in the field on a live plant regardless of it's
type or application. The only variation in the procedure is
what test equipment you will use to calibrate the instrument
with. For instance calibrating a temperature transmitter you
want to take a HART and decade box with you. Calibrating a
pressure , flow or DP transmitter, you want to take a HART
and pressure pump with you like a Druck or hand pump with a
precision gauge on.
Since the transmitter is on line and on a running plant, you
need to determine a couple of things first before you start
working on it. First and most important of all, what does it
do? Is it a control transmitter, a trip transmitter or just
Control transmitter means it's output will go to the DCS and
is linked to a PID controller which will give a continues
output, based on this input, to a control element in the
plant plant like a control valve or a heater element. These
are all critical control components that will trip the plant
if you upset their stability.
Trip transmitter means that it is linked to the ESD system
and will start a plant shut down if it is seeing a to high
or low pressure, flow, DP or level depending on the
application you are working on. Just indication means that
it is completely safe to work on and you cannot trip anything.
To make the control transmitter safe to work on you need to
put the DCS PID controller in manual. This will keep the
output of the controller at the current output position and
bypass the output function of the controller. In a fast
moving application the control room operator will send a
field operator with you to control the control valve
manually with it's hand wheel while you work on the control
To make the trip transmitter safe to work on you need to put
a inhibit in the ESD system output trip function of the
transmitter. This is also sometimes refer to as a MOS
(maintenance override switch).
In both cases the control room operators will do this for
you providing that you have all the permit to work and
inhibit permission paperwork in place.
|Is This Answer Correct ?||16 Yes||0 No|
Answer / guest
throgh HART communicator it is possible to calibrate online
|Is This Answer Correct ?||12 Yes||4 No|
Answer / abdelkader
the first you must see is the transmetter is used as the
loop or indication , if is used as the loop , must avert
operator to put it im manual loop , and if is other case ,
we can calibrate easy without alter the system .Used PID to
determine transmetter function .
|Is This Answer Correct ?||2 Yes||3 No|
what is insdustrial vibration.who it works. give atleast 2 types of insdustrial vibration
what is the formula for changing Range of Transmitter from X mmwc to some Y mmwc with respect to DCS NCMH flow range from x NCMH to y NCMH?
what is commisioning procedure,2)what is is the main work in commisioning in the field tx to dcs, plz send me detail
which gas or which liquid is filled in capillary of dial type temperature indicators on power transformers
how u choose thermowell lenght n pnumatic C/v size?
What is hp & mp seperator .& what is its working principle ?
What is the use of the shield wire and how is the connection at both the ends of a shield wire?
any body tell me ,contol valve has how many cheractristic? explain it.
what is performance test.need defination
sir ,iam in dem urgent need of HPCL Paper of Instrumentation Branch plz send me at your earliest convenience at firstname.lastname@example.org
What are the maximum packet size of: a) HART b) Profibus DP c)Profibus PA d) DeviceNet e) Foundation Fieldbus
What is the difference between PLC and DCS?