why we use 24v dc in instrument rahter than 230vac????
Answers were Sorted based on User's Feedback
Answer / visakh ctjv
all loop powered transmitters are capable of
withstanding 10 t0 36v dc ...as per the process requirement
it will vary.. insome industries they are using 12v dc
also.. 24 vdc is used because as per ISA there are some
standards wich will match other process reqirement like
24vdc relays for digital control also wecan use 24dc to
energize tis relays
|Is This Answer Correct ?||22 Yes||7 No|
Answer / sanjay goswami
24 VDC potential is safe for use in Oil& Gas industries,230
VAC has hight voltage potential and therefore posses high
risk energy for Oil&Gas and human working on it.
DC power is also clean and does not contain any noise like
AC,therefore for precise instrumentation 24 VDC is the
|Is This Answer Correct ?||9 Yes||0 No|
Answer / mohammed abbas
INSTRUMENTS ARE DESIGN TO WORK ON LOW VOLTAGE BY THE
MANUFACTURER LIKE,YOKOGAWA,HONEYWELL AND ROSEMOUNT.LOW DC
VOLTAGE 24 VDC WHICH IS COMPITIBLE FOR ALL THE CONTOL
SYSYTEM AND THE FIELD INSTRUMENT WHICH IS DESIGN WILL WORK
ON 24 VDC WHICH IS REQUIRE TO ENERGIES THE ELECTRONIC PART
ALSO.IF WE USE 230VAC MEANS IT IS DANGER BECAUSE WHICH WILL
CAUSE A ACCIDENT OF LIFE ALSO IF ANY UNSKILLED TECHICIAN
WORK ON THAT INSTRUMENT.24 VDC IS SAFE FOR HUMAN BEING AS
WELL AS LOW POWER CONSUME,WHICH IS COMPITBLE FOR ALL THE
CONTROL SYSTEM AND ALL THE INSTRUMENTS IS BUILD WITH
ELCTROIC WHICH IS FUNCTIONING ON LOW 12 TO 32VDC.ALL THE
INSTRUMENT DESIGN IS ON THE BASIS OF LOW VOLTAGE AND LOW
|Is This Answer Correct ?||12 Yes||5 No|
Answer / narendra mishra
the max voltage is 24v & max current flow throw cable is
27mA so max power loss that cable can bear is 647mW.which
is allowable for the cable designed .
|Is This Answer Correct ?||4 Yes||2 No|
Calibration procedure for electro magnetic flow meter
what is the diffrence between mmwc and kg/cm2 for pressure measurment when we are using mmwc and when kg/cm2 and why????? ?? what is the reason
in my plant 100% feedwater control v/v is air to close but due to instrument air fail,the valve opens but instead of drum level high its get tripped in low low drum level.the urv of level tx is 0 and lrv is -800.pls tell me the reason. for such behaviour.
what is commissioning.
whatis diference between zero pressure and obsolute pressure
why we use impulse line of flow transmitter at 45 deg. angle both side of orifice plate.
can you suggest good traning institute in instrumentation ? (in automation, or designing)
HI FRIENDS IHAVE QUESTION 1.WHAT IS PARTIALSTROKE TEST?WHY IT IS USED ON ESD ON OFF VALVES?
how to make a calliberation for deffirential pressure
Why is the length of cable layyed from RTD to temp. Transmitter should be minimum? Why does heating effect takes place in their?
what is the meaning of GK value(fullform) in flowmeter ?
what is disadvantage of field bus (ff)type of instrument