How can we calibrate THe Auto Titrator ?
Answers were Sorted based on User's Feedback
Answer / chemist
The heart of an autotitrator is the buret itself. You need
to calibrate the buret in terms of dosing rate accuracy.
You should verify at different dosing rate.
|Is This Answer Correct ?||26 Yes||15 No|
Answer / krishnakumar
response of electrode- using ph 4.0 7.0 and 9.0 read the
millivolt reading and make a graph with ph vs mv .graph
slope is between 90 percent to 110percent
burette . by selecting 1 ml 2ml 5 ml 10 ml and 20 ml
titrant is water. check the weight of water in each step
and 1 percent limit is allowed from the selected volume
temperature- compare with calibrated thermometer at minimum
4 intervals .1 percent error is allowed from the calibrater
thermometer.correlation coefficient of temperature graph
actual and set value should be greater than 0.9
|Is This Answer Correct ?||11 Yes||6 No|
Answer / chinnari sanjeevi
The main part of the autitrator is burette.first we calibrate the brunette by using water.after we calculate the disodium tarte water content by using water factor.after we calculate the precision.
|Is This Answer Correct ?||0 Yes||0 No|
Answer / meenakshi sharma
Autoitrator is calibrated using two buffers i.e. pH 4.0 and
7.0 fr two electrodes i.e. aqueous and non aoues , and
limit of slope calculated is -55 to -65 .
|Is This Answer Correct ?||19 Yes||27 No|
can you email me because I have a report and I need you to answer these questions if you would like to if not I completely under stand.
what is the difference between polar covalent bond and hydrogen bond?
father of modern chemistry
why we use only prednisone tablets for disso calibration
what is ceramics
what is meaning of wet analysis.
The end product of anaerobic respiration in human is
how to mesure a relation of assay by HPLC and purity in results?
what is reaction of sodium borohydride and HCl gas..?
hello dears, I am going to make briquttes from coal dust and i want a good binder for it to very cheap and strong if you guys have any edea that helps me pleas answer,
suppose u got 2 or more maxima in the uv spectrophotometer then what u can do
Why sucrose used in calibration of polarimeter