gc Calibration % RSD failed what is the possible reasons?
Answers were Sorted based on User's Feedback
Answer / hiren
First is not column proper installations
Secound is not proper work syrring
Thred is injector septa so much punching
|Is This Answer Correct ?||14 Yes||0 No|
Answer / ahmed
Also because of glass liner damages
|Is This Answer Correct ?||6 Yes||0 No|
Answer / yogesh patel
flow rate of gas not proper its cose fluctuation in assay
|Is This Answer Correct ?||3 Yes||0 No|
Answer / alasyam ramesh
1) Improper stabilisation of column and system.
2) injector septa bleeding
3) partial blockage of detector jet.
4) syring struck due to solvents nature( solvent viscosity).
5) plunger moving and taking improper injection volume.
6) some time may effect room temperature also
|Is This Answer Correct ?||0 Yes||0 No|
What is the difference between the optics of UV and PDA Detectors ?
In Analytical method development related substance method when did follow w/w method when did follow area normalisation method
why 0.02% solution toluene &noHexane using in uv calibration?
what is RI detector Cut-off Value
how much mass should be there in volumetric flask while in preparation of sample for assay?
what is DIPE limit in residual solvent?
How can v calibrate ftir ? Plz explain me
What is viod volume and peak purity in HPLC?
The colour presence in kmno4 is due to.
effect of Ph during analytical run ?
What is the difference between bulk property detectors and Solute property detectors vivimed labs limited
In the isomers,enantiomers give the only physical properties present such that no chemical properties present.