What is the difference between low pressure gredient and high
pressure gredient in hplc instrument?
Answers were Sorted based on User's Feedback
Answer / yojana
In a low pressure gradient pump, different mobile phase is
mixed using a valve before entering the pump head. In a
high pressure gradient pump, each different mobile phase is
delivered by an individual pump head and then mixed at the
pump outlet. The low-pressure gradient design is of lower
cost as it only use one pump head. It can also use more
types of mobile phase without significant increase of cost.
The high-pressure gradient design provides a much faster
gradient since solvent mixing point is much closer to the
column head. This is measured using delay volume. The value
can be 50-300 uL for high pressure gradient pump and can be
2 to 3 times larger for a low pressure gradient pump. A
small delay volume is important when the analysis time is
short or the flow rate is low. If the delay volume is too
large, it become impossible to obtain reproducible gradient
run since the planed composition cannot reach the column
head before a run is finished..
|Is This Answer Correct ?||21 Yes||4 No|
In simple definition,we can say that in low pressure gradient sample mixing with mobile phase will take place before pump and in high pressure gradient it will take place after pump.
Caplin point Ltd-unit-1
|Is This Answer Correct ?||1 Yes||0 No|
Answer / analyticalstudent
|Is This Answer Correct ?||0 Yes||1 No|
Answer / srinivask1985
In Low pressure gradient we are using single pump and one
In High pressure gradient we are using one more pumps and
one more resorvoirs.
|Is This Answer Correct ?||4 Yes||7 No|
what is the difference between potentiometric titration and karl fischer titration?
what is difference between GC & GC-HS
what is the method of UV cabinet CALIBRTION
What is the definition of RRF
which one is better separation in high pressure liquide chromatogram orlowpressure chromato gram in hplc
How to decide restandardisation due or retest or expiry of Volumetric solutions prepared in wet chemistry lab?
How to determine water content of thioanisole? which interfere by KF titration?
Why we are having specification 100% above in case of assay as there is only 100% of compound?
What is cut off value of solvent used in hplc
why lactose has less charges when packed in steel containers when compared with packing of lactose in polyamide material?
difference between HPLC & LC-MS
what is RI detector Cut-off Value