What is the difference between low pressure gredient and high
pressure gredient in hplc instrument?
Answers were Sorted based on User's Feedback
Answer / yojana
In a low pressure gradient pump, different mobile phase is
mixed using a valve before entering the pump head. In a
high pressure gradient pump, each different mobile phase is
delivered by an individual pump head and then mixed at the
pump outlet. The low-pressure gradient design is of lower
cost as it only use one pump head. It can also use more
types of mobile phase without significant increase of cost.
The high-pressure gradient design provides a much faster
gradient since solvent mixing point is much closer to the
column head. This is measured using delay volume. The value
can be 50-300 uL for high pressure gradient pump and can be
2 to 3 times larger for a low pressure gradient pump. A
small delay volume is important when the analysis time is
short or the flow rate is low. If the delay volume is too
large, it become impossible to obtain reproducible gradient
run since the planed composition cannot reach the column
head before a run is finished..
|Is This Answer Correct ?||21 Yes||4 No|
In simple definition,we can say that in low pressure gradient sample mixing with mobile phase will take place before pump and in high pressure gradient it will take place after pump.
Caplin point Ltd-unit-1
|Is This Answer Correct ?||0 Yes||0 No|
Answer / analyticalstudent
|Is This Answer Correct ?||0 Yes||1 No|
Answer / srinivask1985
In Low pressure gradient we are using single pump and one
In High pressure gradient we are using one more pumps and
one more resorvoirs.
|Is This Answer Correct ?||4 Yes||7 No|
difference between AAS & flame photo meter ?
why results are variation in same samples analysis in gc and hplc ?
Is it necessary to inject three replicate for accuracy in AMV.
how ir specrometer is calibrated ?
Which detector is used in uv-vis spectrophotometer?
Through assay by hplc(anhydrous basis) product got 100.8% .How much the potency of that product.Moisture content is 1.7%,Sulfated ash is 0.02%,Total impurities are 0.3%
my name is santhosh i cmpleted b.pharmacy in 2009 then i went to uk in 2010 to pg diploma business management. after i came back to india.. recently again apply to uk for pharmaceutical analysis. they refused my visa you failed to explain y u r change subject. so plz ineed answer for y change the cource
Why we use 100 RPM in friability test
how can we decide the limit of cleaning validation? What is the criteria for it?
principle of UV , PDA , RI detector in HPLC
What is Fourier Transform Spectrosopy?
How to calibrate glassware against std calibrated glassware?