Browse  |   Placement Papers  |   Company  |   Code Snippets  |   Certifications  |   Visa Questions Post Question  |   Post Answer  |   My Panel  |   Search  |   Articles  |   Topics  |   ERRORS new

 Categories >> Engineering >> Instrumentation

 Civil Engineering Interview Questions (3076) Mechanical Engineering Interview Questions (3289) Electrical Engineering Interview Questions (13331) Electronics Communications Interview Questions (2363) Chemical Engineering Interview Questions (416) Aeronautical Engineering Interview Questions (77) Bio Engineering Interview Questions (17) Metallurgy Interview Questions (79) Industrial Engineering Interview Questions (163) Instrumentation Interview Questions (2380) Automobile Engineering Interview Questions (106) Mechatronics Engineering Interview Questions (38) Marine Engineering Interview Questions (43) Power Plant Engineering Interview Questions (77) Engineering AllOther Interview Questions (1196)
Question
```1)HOW CAN CALIBRATE TEMPERATURE TRANSMITTER?
2)HOW CAN CALIBRATE FLOW TRANSMITTER?
3)HOW CAN CALIBRATE LEVEL TRANSMITTER?```
Question Submitted By :: Instrumentation
I also faced this Question!!     Answer Posted By

# 1
1 prepare the instruments needed in calibrating temp.
transmetter.
2 unsrew the resistance input of the transmetter, and
connect it to a resistance box.
3. unsrew the output side of the transmetter, connect the
ammeter serries with the supply of 24 volts dc.
4. supply it with the resistance of 0% or 100 Ohm until
100%. and record it to the data in the as found.
5. the next is the zeroing. supply the input with 100ohm
twisting slowly to the left to have an output of 4mA.
6. span- supply the transmitter with 100% depends on the
value of high temperature of the RTD value. and adjust it
to 20mA.
7 repeat it up to 4-5 times.
then supply is again with 100Ohn, mast have an output of
4mA until 100%.
record it to the data as asleft.
for 0% must have an output of 4mA, 25% 8mA, 50% 12mA, 75%
16Ma, and 20mA for 100%.

 Is This Answer Correct ? 77 Yes 12 No
Tontoy

# 2
1)There is equipment available where we can put RTD and
give temperature.Now set temp transmitter parameters to
produce mA equivalent to temp given.

 Is This Answer Correct ? 67 Yes 23 No

# 3
1. First select the calibrating equipment
2. for ex: For RTD, select RTD circuit with zero and span adjustment potentiometers.
3. make the input quantity (temperature) to zero.
4. adjust the zero adjustment pot ntill the output of the RTD becomes zero.
5. make the input quantity to high (Ex.100 deg) and adjust the span pot till the output becomes 100.
6. Repeat steps 3 through 5 till there are no changes in zero and span pot.
7. now the instrument is calibrated.

 Is This Answer Correct ? 9 Yes 18 No
Gcs

  Question Asked @ Answers Other Instrumentation Interview Questions what is the relation between flow and pressure.for example relationship between level and pressure is p = phg like that i want ans for this question.and also how to calculate orifice sizing and flow transmitter. pls some one ans for this questions. Orchid 3 how to measure low temperature(<100 dec celsisus) with non condact sensor 2 While preparing the graphic diagrams trend display, 3 types of trends are prepared by Yokogawa system vender. These are for process variable, manipulated variable & set point variable. Why these 3 trends are required? Explain the purpose of these 3 trends. 1 what is the diffrence between mmwc and kg/cm2 for pressure measurment when we are using mmwc and when kg/cm2 and why????? ?? what is the reason Jindal 11 explane hart protocol? what is its advantages. Jindal 3 is any standard angle for thermowell mounting for horizontal and vertical pipe lines? 3 WHAT IS D/O D/I AND A/O A/I PLZ EXPLAIN 2 1.Give example of analog and digital signal 2. What is yhe advantage of 8086 than 8085 Bhel 5 tell me about the instrumentation,what is the actual need of it ?what are the characteristics of a IC engineer ? L&T 11 I want to know if flowrate increases in 16' pipe line what will happen to upstream pressure and why? 1 if a tank is contaning oil and water than how can we measure the oil and water level seperately 3 Troubleshoot to the device level basic process control systems given the Process and Instrumentation Diagram? HPCL 1 For more Instrumentation Interview Questions Click Here