why any measurement made interms of pressure is more
to be more precise
most of the flow velocity is made interms of pressure and it
what is the reason behind pressure measurement being
My rtd working well at the same time r/i converter is not working properly.when power is given to rtd..it is giving 30mA and getting 2.4 mA if i remove the compensate leads No matter how many ohms.why it is happening like so?what is the solution?rtd range-0-200deg c& 4-20 mA o/p.please give me solutiin anyone......
1.In which princeple DP transmeter works?
2.Why UPS system is use for DCS?
3.How drum lvel controller works?
4.What is the difference between glob valve and gate valve?
5.How axial shift probe work & in which princeple it works?
How to work togather like sov,limit switch,proxy switch
and booster in xv
Dear all... I have 1 0-400 duplex type RTD, I connected the
A side to the temperature indicator and B side to the PLC
panel, but when I have observed then found that The PLC side
temp has the variation. and it was distub to other RTD temp
also. so what can I do.